Troubleshooting and User Guides
The walk-through video of how to connect your Piranha (Red or Black) via Bluetooth to your windows laptop PC or tablet, is found here.
Windows may sometimes block Bluetooth devices from being found. If Ocean fails to get your Piranha wireless connected to your Windows 10 Tablet or Laptop, a new pairing in Windows is then required.
The workaround of how to connect your Piranha when you have a problem using Bluetooth and Windows 10 together with the Black Piranha is found here.
Application and measurements
There are a number of ways to do this with Ocean software. One way is to choose TIMED MODE and make a single measurement to include all the exposures. The result will be your accumulated value.
The second option allows the user to make any number of measurements and simply add the values at the end of the exposure sequence.
Yes, and it is standard on all Piranha 500 and 600 systems. This calibration has been available for RTI products since 2007.
By using the 1 point method you only need one exposure in the center hole for each CTDIw estimation. This is possible by using know relations in the dose distribution between the center hole and the four periphery holes. This relation is defined by the “k-factor”.
Just making one exposure instead of five saves both time and the X-ray equipment.
The k-factors have been extracted from statistics for many years from measurements of the CTDIw and CTDI100 measured with ion chambers on different scanners.
The variation between different scanners and bowtie filters is less than ±5%. This difference gives less distribution to the expanded uncertainty than influences by measuring the five periphery holes individually.
Factors that add inaccuracy when measuring in the periphery holes are energy dependencies, low signal, and high dynamic variation in signal during the tube rotation.
When using the CT dose profiler in the periphery holes during a scan, the pitch has to be less than 0,5 to avoid loss of information for the dose estimation.
Yes. You can easily extract your k-factor for each of your CT scanners by using previous measurements with ion chambers.
RTI strongly recommends the 1 point method to be used for CTDIw measurements with the CT Dose Profiler. The method saves time and gives higher accuracy than the 5 point method, either using a pencil ion chamber or the CT Dose Profiler.
Yes, the Piranha can, depending on the model, measure on all different kinds of X-ray equipment. But there are also simpler models available at a lower cost.
Yes, by using the RTI Chamber Adapter you can connect the ion chambers to the Piranha. Please check with us to make sure that the ion chamber you intend to use is fully suited for the chamber adapter.
Yes, all measurements made with the Piranha can be saved with the Ocean software. It allows the user to create templates for easy measurements. For more information, see our Software page.
Yes, the Piranha 657 can measure dental/panoramic units. The user simply selects the Dental or Panoramic option in the display.
When using the “Position Check” function you are sure that the Piranha is accurately placed and the Piranha detector area is uniformly radiated. For easy and accurate Panoramic measurements – use our optional “Panoramic Dental Holder”.
Measured values can include kVp, time, dose, dose rate, TF, and HVL. Waveforms are shown instantly. Dental measurement capabilities are standard on all 200, 500, and 600 series Piranhas.
Yes, any RTI MAS Probe can be used simultaneously with the Piranha during X-ray exposure, making it possible to display up to 8 parameters at one time: kVp, time, dose, dose rate, TF, HVL, mA, and mAs. Waveforms for kV, dose, and mA waveforms are available with each exposure.
The Piranha can measure the correct kV on very low dose rates and for the whole kV spectrum. For example, the Piranha can measure 43 kV at a dose rate of 30 mR/min at 25 µA consistently and with no special set-up.
The Piranha has multiple solid-state detectors that measure simultaneously during any exposure. Some of these detectors are dedicated to measuring kV and Total Filtration (TF). The correct HVL is calculated by using the measured kV and TF values. HVL in a single exposure is available for R/F, Mammo, Dental, and CT.
The user can also measure HVL the conventional method using aluminum sheets. Simple, ready-to-use HVL applications are included in both Ocean and QA-Browser software.
Yes, the Piranha can measure a large variety of parameters (kV, dose, time, HVL, total filtration, frames, dose rate, frame rate, etc.) – all in just ONE exposure!
No, RTI uses Bluetooth in our instruments. This eliminates unnecessary cables and makes the setup easier and faster.
All RTI instruments are battery powered – so no external power supply needed.
A fully charged Piranha can be used for up to 15 hours, and can be charged and used simultaneously. The Piranha can be charged via a regular wall outlet or the supplied USB cable connected to a laptop.
Simply choose “Fluro and Frame-rate” with Ocean software or “Pulsed Fluoro” with the Piranha QA Browser. This will work with both the Piranha and/or its external dose detector.
With Ocean software, you can also measure a lot of other frame parameters like total frame count, dose/frame, pulse width, etc.
Yes. The Piranha can measure as low as 30 mR/min (or even lower). Basically, as long as the Piranha can measure the dose rate, the Piranha measures correct kV.
All Red Piranhas can be upgraded to full Black Piranha functionality, including the new Black shell. Please contact your local distributor for more information and current upgrade prices.
The Cobia Family
The Cobia can be positioned in any direction, as long as the X-ray field covers the whole detector area.
RTI software is optimized for Windows 7, 8, and 10 but it is also possible to use Windows Vista.
Service and repairs
Start by filling out our Service Order Form (available here). Here you tell us what you want to send into us. After that, please contact our distributor for your country, they will guide you in your choice of calibrations, repairs, and upgrades (Sometimes you can find answers to your problems in the manual or by updating to the latest versions of software and/or firmware).
If we don’t have a distributor for your country, please contact RTI directly.
RTI usually has two software/firmware releases each year. In each release, new features are presented which will simplify the measurements and/or the analysis and handling of them. These updates can be downloaded for free from the website Downloads page.
RTI recommends a recalibration after two years.
RTI has an average turn-around of two weeks.
Yes, always contact us or your local distributor before sending in your equipment.
No, not yet.
Training & Seminars
Yes, we give demonstrations while traveling around the world to participate at different trade shows and exhibitions. It is also possible to contact us and get a private demonstration of our products and software. Contact Erik Wikström, Manager Training, about the RTI Academy.